*With 30-second time frame.
†CONTOUR® NEXT LINK 2.4 meter meets ±10% accuracy vs. laboratory method for blood glucose concentrations, specifically: 99.3% of results within ±10% for blood glucose concentrations ≥5.55 mmol/L, and 98.4% of results within ±0.56 mmol/L accuracy vs. laboratory method for blood glucose concentrations <5.55 mmol/L.
Insert the grey end of a CONTOUR® NEXT test strip into the meter.
Lightly touch the test strip to the drop of blood and hold until the meter beeps. The test result will be displayed after a 5-second countdown
Send your results to your Medtronic MiniMedTM 630G pump.
Sending test results wirelessly to your pump
There are two easy ways to connect your meter to your pump: Manual Connect and Auto Connect. Use Auto Connect to quickly and easily connect your meter to your pump. Use Manual Connect if there are other Medtronic devices nearby. With Manual Connect, you will be prompted to enter the device (meter) serial number so the pump can find your meter. With Auto Connect, there is no need to enter a serial number. The meter and pump search for each other, and you just need to confirm the serial numbers to connect. For more information on linking your meter and pump, refer to the CONTOUR® NEXT LINK 2.4 User Guide or Quick Reference Guide that came with your meter.